MEDICAL TREATMENT SET FOR OBTAINING PLATELET-RICH PLASMA
Thanks to its repair and regenerative properties, it allows you to eliminate fine and deep wrinkles, acne scars and stretch marks, sun damage, shadows, and puffiness under the eyes. The treatment helps to regain the lost face volume as well as improve its tone and structure. Thanks to the stimulation of angiogenesis, the skin is better supplied with blood and oxygenated, which results in a significant improvement in colour and functionality. It was also found to have a positive effect on the stimulation of the hair follicles and hair growth.
Tubes are intended for reimplantation procedures and are marked with a medical certificate of group II B in vivo. Tubes allow a vacuum, completely closed blood collection system, which increases the safety of the procedure and prevents material contamination. Filled with ACD anticoagulant solution (sodium citrate, citric acid, glucose), preventing premature clotting of the collected blood. Platelets are naturally activated by glucose only after reimplantation into tissues. The production of medical polycarbonate tubes results in the maximization of the level of platelets in the PRP concentrate and an increase in the effectiveness of the therapy. Each of the tubes is packed in a separate blister, which allows the set to be divided into 2 independent treatments.
2-treatment sets with treatment instruments
ACD 9ml / 15ml tube
The kit includes 2 polycarbonate, sterile, vacuum tubes with a capacity of 9 ml, filled with the ACD anticoagulant solution (sodium citrate, citric acid, glucose), preventing premature clotting of the collected blood. After-care bags allow a vacuum, completely closed blood collection system, which increases the safety of the procedure and prevents contamination of the material. The production of medical polycarbonate tubes results in the maximization of the level of platelets (PLT) in the PRP concentrate and an increase in the effectiveness of the therapy. Each of the tubes is packed in a separate blister, which allows the set to be divided into 2 independent treatments. Tubes are intended for reimplantation procedures and are marked with a medical certificate of group II B. Additionally, the kit includes 2 disposable treatment equipment sets containing:
– butterfly needle with a tubing
– system needle for vacuum blood collection 21G 0.8x30mm
– collection needle PRP 20G 0.9x70mm
– a luer-lock syringe with a capacity of 3 ml
– luer-lock syringe with a capacity of 5 ml
– 2 injection needles 30G 0.3x13mm
– 2 injection needles 32G 0.23x4mm
HOW TO USE:
Stage 1: Collection of Whole Blood
Before starting blood sampling, disinfect the skin. Open the set of treatment instruments and the blister with the tube. Make a venous puncture with the included phlebotomy accessories, then connect the tube with the holder. The test tube will begin to automatically fill with blood to the volume of about 9ml. After filling, disconnect the tube and, if necessary, attach another. Turn the received test tubes upside down several times in order to mix the blood with the ACD solution. Place contaminated collection instruments in a medical waste container and dispose of according to procedures for handling blood-contaminated materials.
Stage 2: Centrifugation
Place the received test tube in a centrifuge. On the opposite side, sweep a counterweight filled with water or another blood tube to balance the device. Centrifuge for 5 minutes at 1100g centrifugal force (RCF). Set the rotational speed according to the recommendations of the manufacturers of the centrifuge.
Stage 3: PRP Collection
The effect of centrifugal force will arrange the morphotic parts according to the gravitational weight and obtain 3 separate fractions. There will be red blood cells at the bottom of the test tube, platelets (PLT) above them, and platelet poor plasma (PPP) above them. Use a 20G 0.9x70mm needle and draw the PLT plate concentrate (in circular movements) directly into the syringe (approx. 0.5-0.7 ml). Then select about 2-2.5 ml of platelet poor plasma (PPP) and mix everything gently. From the separation of 1 tube, you will receive a total of about 2.5-3.2 ml of platelet-rich plasma (PRP).
Stage 4: Injection
The resulting PRP solution is immediately applied to the tissues. The administration of PRP within 30 minutes from the moment of collection ensures full biological activity of the solution. It is possible to use some or all of the obtained preparation.
Stage 5: Disposal
After use, dispose of all components of the procedure set in accordance with the guidelines for handling materials potentially contaminated with blood. Use caution from unprotected needles and other contaminated treatment instruments.
DOSAGE AND INJECTION METHOD:
Joint space injections should be performed with medical imaging and with the same precautions as for any other intra-articular injection. Immediately after the injection, the patient should refrain from any activities that weigh on the joints. Intensive physical activity should be limited within 48 hours after the procedure. To obtain optimal treatment effects, it is recommended to use 1 treatment tube per joint.
Perform intradermal injections using aseptic techniques and precautions. The obtained PRP concentrate should be implanted directly into the deep layers of the dermis with the use of needle mesotherapy using the nappage or micro-deposit technique, and in places requiring special regeneration using the retrograde linear technique. To obtain optimal treatment effects, it is recommended to use 1 tube per treatment area.
The external use of the PRP plate concentrate is possible in conjunction with other aesthetic medicine procedures (e.g., microneedle mesotherapy, fractional laser). Apply the obtained product epidermally after damaging the skin.
Injections of PRP concentrate may cause redness, swelling, pain or slight bruising at the treatment site. Usually, these changes are mild and disappear after a few days. To obtain optimal correction effects and safe convalescence:
– do not apply makeup for up to 12 hours
– avoid solarium and excessive sun exposure for up to 2 weeks after the procedure
– do not expose the correction site to extreme low and high temperatures
– apply ointments that seal blood vessels to the place where the bruising occurred
– immediately report any disturbing changes to the specialist performing the procedure
DIVES MED PRP may only be used by doctors and medical personnel with postgraduate training in the techniques of using autologous blood products. Before use, familiarize yourself with the purpose and method of use of the set. Before starting the procedure, it is absolutely necessary to conduct a medical history, rule out contraindications, obtain informed consent to perform the procedure and inform the patient about possible results and possible side effects. During the implantation of the product, the procedure and the injection site should be fully aseptic. Do not inject intravenously or use for other purposes. Single-use test tube and instruments. Do not re-use and do not re-sterilize. Do not use after the expiration date. Incorrect or inconsistent with the manufacturer’s recommendations may lead to complications and side effects.
POSSIBLE SIDE EFFECTS:
DIVES MED PRP ACD injections may lead to slight damage to blood vessels and the formation of hematomas and bruises. Immediately after injection, the treatment area may experience swelling and redness, as well as a short-term secondary inflammatory reaction.
The DIVES MED PRP ACD set contains 2 treatment sets for single use. Dispose of all used surgical instruments in a container for medical waste. After use, dispose of all parts of the set according to instructions for the handling of potentially blood-contaminated materials. Use caution from unguarded needles and other contaminated surgical instruments. Do not put used needles back into the needle cap.
The contents of the blisters have been sterilized by radiation.
SHELF LIFE AND STORAGE:
Store at 2oC-28oC, away from sunlight and heat sources. Do not use the product after the expiry date and if the blister is damaged or has been incorrectly stored. Use the content of the blister and the treatment instruments for a single application and do not re-sterilize.